Patient Sequence No: 1, Text Type: N, H10
It was determined that the signal loss was related to the mobile application. It was indicated that the operating system for the smart device was not a supported system, which is off-label usage of the device. Data was provided for evaluation. Per (b)(4) cases where the sensor session line is missing in share support tool and share support service, complaints will be closed since a data investigation cannot be completed as data ambiguity cannot be resolved further. The allegation was undetermined. The probable cause could not be determined. No injury or medical intervention was reported.
Patient Sequence No: 1, Text Type: D, B5